ISO. Second edition. Reference number Permission can be requested from either ISO at the address below or ISO’s member body in the. 31 Jul Purchase your copy of BS EN ISO as a PDF download or hard copy directly from the official BSI Shop. All BSI British Standards. EN ISO Intravascular catheters – Sterile and single-use catheters – Part 5 : Over-needle peripheral catheters – ISO specifies requirements.
|Published (Last):||5 July 2011|
|PDF File Size:||10.51 Mb|
|ePub File Size:||14.97 Mb|
|Price:||Free* [*Free Regsitration Required]|
No part of the these publications may be reproduced in any form without the prior permission in writing of BIS.
BS EN ISO 10555-5:2013
Copyright BIS has the copyright iso 10555-5 all its publications. Standards are also reviewed periodically; a standard alongwith amendments is reaffirmed when such review indicates that no changes are needed; if the review indicates that changes are needed, it is taken up for revision.
Assembly comprising the cath- eter tube, catheter hub and any integral fittings. Attention is particularly drawn to the following: The fluid iso 10555-5 in the needle shall be free of unintended 1055-5 that would prevent flashback.
An approved test method for producing a value 1055-5 radio-detectability will be established. Lso of Indian Standards BIS is a statutory institution established under the Bureau of Iso 10555-5 Standards Act, to promote harmonious development of the activities of standardization, marking and quality certification of goods and attending to connected matters in the country.
Annex D shows examples of typical iso 10555-5 point geom- etries. Catheter designed for the introduction or withdrawal of liquids or devices into or from the peripheral vascular system.
Intravascular catheters ISO , -3, -4, -5
The number iso 10555-5 significant iso 10555-5 retained in the rounded off value should be the same as that of the specified value in this standard. The assembly comprising at least a needle tube attached to and communicating with a needle isl.
The corresponding Indian Standards which are to be substituted in their places are listed below along with their degree of equivalence for the editions indicated: NOTE To improve the visibility of the solution, a colorant such as red or blue food dye may be incorporated. I Principle A force is applied successively tensile and connpress- ive to the needle tube iso 10555-5 needle hub and the tube- hub union is then examined for loosening.
If a steel tube is used, it shall connply with ISO Round the calculated average water fiowrate to the nearest whole number of millilitres. In this adopted standard, reference appears to certain International Standards for which Indian Standards also exist. Available from 10555-55 Color, N. Assembly comprising at least a needle tube attached to, and communicating with, a iso 10555-5 hub.
Certain conventions are, however, not identical to those used in Indian Standards. Limited collection of colours.
Details of which is given in National Annex A. Isi iso 10555-5 The point iso 10555-5 be designed to be noncoring. When tested in accordance with annex E, fluid shall not leak out of the vent fitting within 15 s. Suggested colour references for opaque naaterials are given in annex C.
Peripheral intravascular catheter designed for the introduction of withdrawal of liquids or devices into or from the lso vascular system. The fluid is allowed to flow into the needle, and the time taken for fluid to leak through the vent iso 10555-5 is measured. Search the history of over billion web pages on iso 10555-5 Internet.
This does not preclude the free use, in the course of implementing the standard, of necessary details, such as symbols and sizes, type or grade designations. Until that time, 10555–5 manufacturer may label his product “radio- opaque” provided he can support this claim by demonstrat- ing that he has an appropriate method for showing iso 10555-5 opacity. For the purpose of deciding whether a particular requirement of this standard is complied with, the final value, observed or calculated expressing the result of a test or analysis, shall be rounded off in accordance with IS 2: The catheter tube may have ixo single lumen or multiple lumens.
An example of a suitable apparatus is shown in fig- ure B. Add the following reference: At the time of publication, the editions indicated were valid. Pages 1 and 2 4. When the needle is fully inserted into the catheter unit, the catheter tube shall neither extend beyond the heel of the needle iso 10555-5 nor be nnore than 1 nnnn fronn it see dinnension a in figure 1.
The details of conditions under which the licence for the use of the Standard Iso 10555-5 may iso 10555-5 granted to manufacturers or producers may be obtained from the Bureau of Indian Standards.
Iso 10555-5 1 — Typical iso 10555-5 peripheral intravascular catheter NOTE 2 At the time of iso 10555-5 of this part of ISOthere is no acceptable, validated test method to determine radio-detectability.
Is tube with one end sharpened to facilitate entry into body tissue. Review of Indian Standards Amendments are issued to standards iso 10555-5 the need arises on the basis of comments.
iso 10555-5 Fitting attached to the needle tube, providing communication with its bore. I Principle Water is allowed to flow through 15055-5 catheter and the amount of flow is measured either volumetrically or gravimetrically.